Constructional design / case
|Mechanical design:||measuring insert with connection socket per DIN 43735|
|measuring insert: stainless steel mat.-no. 1.4571 (316 Ti)|
|length and Ø see order details.
Connection socket Ø 42, max. 6 terminals or prepared for transmitter mounting
|Measuring resistor:||measuring resistor Pt 100 per DIN 60751
nominal value of Pt 100 sensor: 100 Ohm at 0 °C
|Application:||-200 to max. 850 °C
standard: -50...+400 °C
see order details
|Sensor limiting error:
|the limiting error is defined for classes: see data sheet|
|Measuring insert length:||Please select standard lengths. The length of the measuring insert is to be selected so that the measuring insert stands on the thermowell bottom. This ensures good heat transfer. The measu-ring inserts are spring loaded (spring travel: max. 10 mm) to ensure that they are pressed down on the bottom of the thermowell. We recommend the use of thermolube.
Standard lengths see order details.
Special lengths are possible.
|Mounting of transmitter:||Pt 100 transmitter for head mounting can be mounted instead of terminal socket.|
Tests and Certificates
|Ex-approval:||BVS 04 ATEX E 144 X
II 2G Ex ia IIC T4/T6
Ui ≤ 30 V
Pi ≤ 200 mW
|Functional safety:||per EN 61508, classification per SIL2;
without transmitter, only